Masimo rainbow Acoustic Monitoring provides similar respiration rate accuracy as capnography respiration rate monitoring 1
|Dataset||Number of Samples||Bias (brpm)||Standard Deviation (brpm)||Root Mean Square Accuracy (brpm)|
|Capnography Respiration Rate||21,405||0.22||1.62||1.63|
- > Both methods were compared to a control respiration rate obtained by a trained observer counting inspirations and expirations visually and by listening (as done during auscultation).
- 1 Masimo FDA Submission Data.